After completion of the automatic power-on test, the status of the balance operating program appears in the weight
display: L stands for the list mode. In this mode, you can only verify the code setting, but you cannot program new
codes. If you want to change a program code, you must first unlock the program access.
To do so, slide the unlocking switch (10) at the forward right of your balance in the arrow direction.
The display will signal C representing the change mode, and you can now proceed to make the necessary code
After the balance operating program has been called, the display will show a continuous numerical sequence from 0
to 5 representing the page selection, in addition to the status signal L or C.
When your selected number for the
page appears, push the tare bar. The page code number is now fixed in the display, and the cycle for the line
starts. Again confirm your selected number with the tare bar, and your selection will be fixed. Next the word cycle
When the 0symbol apears, this marks the actual setting.
To make changes (C mode), press the tare bar when the appropriate code appears.
Brief display of BUSY and the osymbol confirms your selection, followed by return to the zero representing the
To return to the weighing program:
push the tare bar each time a 0 appears in the numerical cycle (word, line, page). If you have made code changes,
your code entry is stored as soon as the display returns to the weighing mode. Lock the balance program with the
menu access switch (display L) and replace the protective cap.
- 8 -